Throughout this manual, the following conventions are used for SCPI command syntax. • Square brackets .. HEWLETT-PACKARD,EA,0,X.X-X.X-X.X. Read about ‘EA Triple Output DC Power Supply User’s Guide’ on element14 .com. EA Triple Output DC Power Supply User’s Guide. Items are supplied with manuals, accessories and typically a full no-quibble 2 . HP EA/32A/33A/34A Programmable DC Power Supply Specifications.
|Published (Last):||5 August 2017|
|PDF File Size:||13.67 Mb|
|ePub File Size:||9.89 Mb|
|Price:||Free* [*Free Regsitration Required]|
Contact the seller – opens in a new window or tab and request a shipping method to your location.
I would suggest you use the instrument. Read more about the condition. When ordering from the US, parcels may be subject to import tax and duty charges, which the buyer is responsible to pay. Learn more – opens in new window or tab. Sorry if I my my post was a little e3631z confusing. Figure shows a controlled transformer tap as used in the Agilent EA.
HP EA Triple Output DC Power Supply Service Manual E B-3 | eBay
I am able to control almost every settings of the power supply voltage at each channel, limit current at each channeletc. Remove the fuse-holder 2 Install the correct line fuse.
Each supply sends back to the control circuit voltages representing outputs at the terminals. Contact the manuql – opens in a new window or tab and request a shipping method to your location. Back to home page. HP A Function Generator. Output Voltage Overshoot During turn-on or turn-off of ac power, output plus overshoot will not exceed 1 V if the output control is set to less than 1 V.
Hewlett Packard EA Triple Output DC Power Supply HP 6v 5a | eBay
Skip to main content. Page 85 The command returns a string with the following format be sure to dimension a string variable with at least 40 characters: Estimated on or before Tue. That terminal must be connected to earth ground through a power source with a 3-wire ground receptacle. Both of these are of very low value in the Agilent EA. This program uses the status registers to generate an 30!
Instrument Control (GPIB, Serial, VISA, IVI)
The most important gp limiting the speed of output voltage change is the output capacitor and load resistor. Seller information saradean3 For additional information, see the Global Shipping Program terms and conditions – opens in a new window or tab This amount includes applicable customs duties, taxes, brokerage and other fees.
Add to watch list Remove from watch list. Because of the special behavior of my power supply the called function only returns a non error value.
Hewlett Packard E3631A Triple Output DC Power Supply HP 6v 5a
E33631a in to check out Check out as guest. Interest will be charged to your account from the purchase date if the balance is not paid in full within 6 months.
manal This command queries the power supply to determine the number of times it has been calibrated. The item may have some signs of cosmetic wear, but is fully operational and functions as intended.
Be sure to dimension a string variable with at least 40 characters. The item may have some signs of cosmetic wear, but is fully operational and functions as intended.
Even if the cable has the proper connectors for your system, the internal wiring may be incorrect. Remote Programming Chapter 7 Tutorial Remote Programming Remote Programming During remote programming a constant-voltage e361a power supply is called upon to change its output voltage rapidly.
Your power supply is equipped with a 3-wire grounding type power cord; Email to friends Share on Facebook – opens in a new window or tab Share on Twitter – opens in a new window or tab Share on Pinterest – opens in a new window or tab Add to watch list.
For each step, use the keys shown on the left margins. The item you’ve selected wasn’t added to your cart. The calibration code may contain up to 12 characters over the remote interface.
Back to home page Return to top. However, I cannot enable the output.